I have a well adjusted calcium reactor and last night I changed my reagent. I ran two tests and it seemed to be testing as it was prior to the refill. I went to bed and when I checked my app in the morning the alkatronic had tested twice at a much lower dkh. Because of this, it dosed 74 ml of alk solution. This caused my dkh to jump over 1 full point. I sure hope my sps don't stn from this! There really needs to be an ability for me to say never dose over 20 ml or whatever value I might choose. I know with my calcium reactor I would never ever need 74 ml! Now I am scared to use action mode. Please please update this!
Posts made by David Johnson
RE: Suggestionsposted in Alkatronic •